Agilent Varian ConvecTorr Convection-PType Gauge Simulator and Reference Tube,
Agilent pn: L7382301
The Convectorr Simulator (L7382-301 P type) is intended to be a convenient tool for verifying proper operation of the MultiGauge, SennTorr or XGS-600 control units. For those cases when the target
gauge tube can not be brought to proper pressure levels for true in-situ calibration, the simulator can
be used to at least insure the control unit is calibrated to the average tube.
Normally, the simulator would be plugged into the target gauge cable at the sensor end. When the
switch is set to ATM the controller should read approximately 1 x 105 Pa (760 Torr, 1000mBar).
If desired, the controller may be "calibrated" by pressing the (ATM) button on MultiGauge or the
(CAL) button on SenTorr or CAL on main screen on the XGS-600. Sliding the switch to VAC should cause the controller to read approximately
1.33 x 10-1 Pa (1 mT, 1.3 x 10-3 mBar). Again, the controller may be "calibrated" by pressing the appropriate
button on the controller keypad.
The Simulator is not intended to take the place of NIST-traceable calibration.
The product is intended to verify the functionality of control electronics.
For those situations requiring NIST-traceable calibration, Agilent Varian
offers the comprehensive STARRS P Program.
Can be used with the following controllers:
- Varian Multi-Gauge Controller
- Varian Senn Torr Gauge Controller
- Varian XGS 600 Gauge Controller
This is a Separate ConvecTorr Gauge Simulator Only, Cables and Controllers available on this website.